BRYMEN BM859CF PDF
BMs – Top class digital multimeter made by Brymen. counts (5 4/5 digit), High Performance, User-oriented innovative features. Unbeatable 1µV. USER’S MANUAL BM BMCF 1 1) SAFETY This manual contains .. BRYMEN’s warranty does not apply to accessories, fuses, fusible resistors, spark . Thanks Lightages: + didn’t know TME also ships to the Netherlands prices are looking good indeed and no problem with customs since it’s.
|Published (Last):||21 September 2013|
|PDF File Size:||7.22 Mb|
|ePub File Size:||6.5 Mb|
|Price:||Free* [*Free Regsitration Required]|
Cleaning and Storage Periodically wipe the case with a damp cloth and mild detergent; do not use abrasives or solvents. Press the button momentarily again to step through the ranges. Install only the same type of fuse or equivalent Trouble Shooting If the instrument fails to operate, check battery, fuses, leads, etc. Press button momentarily to toggle between 5 full digits and 6 full digits Hz readings. These voltage levels pose a potential shock hazard to the user.
Continuity function is convenient for checking wiring connections and operation of switches. When activated from voltage function, you can also press the RANGE button momentarily to select rbymen trigger level range manually.
It is excellent in detecting faulty contacts, identifying potentiometer clicks, and indicating signal spikes during adjustments. BRYMEN’s warranty does not apply to accessories, fuses, fusible resistors, spark gaps, batteries or any product which, in BRYMEN’s opinion, has been misused, altered, neglected, or bmm859cf by accident or abnormal conditions of operation or handling.
Inspect test leads, connectors, and probes for damaged insulation or exposed metal before using the instrument. In many cases the suspected component must be disconnected from the circuit to obtain an accurate reading. Hold The hold function freezes the display for later view. If the Hz reading is unstable, select lower sensitivity to avoid electrical noise. Manual ranging mode feature is not available in Hz function. A full wave rectified sine waveform is a good example, and the AC true RMS function will only give the AC component reading which is at Loosen the 4 screws from the case bottom.
The digital display shows OL if the diode is good. Normal forward voltage drop forward biased for a good silicon diode is between 0. Insert the banana plug B8m59cf temperature bead probe Bkp60 standard accessory x 1 with correct polarities.
Bryemn to the explanation in this Manual!
Brymen BMCF Manuals
An OL indicates an open diode defective. The new setting will be saved automatically to the non-volatile memory as power up default. Press and hold the button for 1 second or more to resume auto-ranging mode.
Do not measure any current that exceeds the current rating of the protection fuse.
The analog bar-graph remains the brumen current value to alert the user. Most measuring functions through this terminal will then be open circuit. Dual channels T1-T2 readings require 2 probes. It’s a community-based project which helps to repair anything.
Always turn the rotary switch to the OFF position when the meter is not in use. Do not attempt a current measurement to any circuit bm859c the open circuit voltage is above the protection fuse voltage rating.
Auto Power Off feature will be disabled automatically in this mode. Line Level Frequency measuring function input sensitivity varies automatically with voltage or current function range selected. Note — Examples include electricity meters and primary over-current protection equipment.
If the reading shows zero, select higher sensitivity. The analog bargraph pointer will point at the selected trigger level range scale 1, 2, 3, or 4. In voltage or current functions, press the Hz push button momentarily to activate or to exit Line Level Frequency measuring function. Suspected open circuit voltage should be checked with voltage functions.
Brymen BM859CF Manuals
RSC PC computer interface capabilities The brgmen equips with an optical isolated interface port at the meter back for data communication. The lower the measuring range the higher the sensitivity.
That is, mV function has the highest and the V range has the lowest as in voltage function ranges. Line Level Frequency measuring function is designed especially for noisy electrical high voltage signals. You can also use a plug adapter Bkb32 Optional purchase with banana pins to K-type socket to adapt other standard K type mini plug temperature probes.