ABBOTT CELL DYN SAPPHIRE PDF
Korean J Lab Med. Jun;27(3) [Evaluation of the Abbott Cell-Dyn Sapphire hematology analyzer]. [Article in Korean]. Park Y(1), Song J, Song S. Name of instrument, CELL-DYN Sapphire*. First year installed in U.S./Outside U.S./No. of units sold in , //—. No. units installed in U.S./Outside. The CELL-DYN Sapphire hematology system is designed to address routine testing needs of high-volume hematology laboratories and difficult.
|Published (Last):||9 February 2017|
|PDF File Size:||3.63 Mb|
|ePub File Size:||12.6 Mb|
|Price:||Free* [*Free Regsitration Required]|
Inquire Here Auto Sample Handling: For assessing the differential count, the percentages dynn lymphocytes and monocytes determined by the Cell-Dyn Sapphire were added, and this sum was compared with the mononuclear cell MNC count found on the cytocentrifuged preparations, which included lymphocytes, monocytes, and, in some samples, epithelial cells and macrophages.
Cell Dyn Sapphire – Abbott – Hematology Analyzer
But more interestingly, a screening opportunity exists within the very good sensitivity of The names and logos of manufacturers, their instruments, and their products referred to herein may be protected by ssapphire or other law, and are used herein solely for purpose of reference.
For method comparison, we used CSF, serous fluid 61 pleural and 40 peritonealand CAPD fluid samples that were collected for routine diagnostic purposes.
First, all requested tests were performed. A P value of less than. The analyses were performed as soon as possible and for CSF, always within 1 hour from the time of sample collection.
Why request a quote through SelectScience? Inquire Here Power Supply: Inquire Here Optical System: For CSF specimens, many cases were falsely classified as abnormal, leading to a very poor specificity of They include ecll automated alarm systems for indicating results outside the reference range.
Sapphide Cell-Dyn Sapphire flagged 7 samples, but in only 2 of those samples were metastatic cells present according to the differentiation of the cytocentrifuged preparation. Inquire Here Reagent System: The positive and negative likelihood were respectively 3.
Cell-Dyn from Cell Dyn Sapphire. Wapphire counts are important for the diagnosis of intracerebral hemorrhage and for the exclusion of a traumatic tap as the cause of an elevated WBC count.
CELL-DYN Sapphire Hematology Analyzer from Abbott | SelectScience
MAPSS plus 3 color fluorescent technology. We evaluated the out of bound greater than preset limit flagging efficiency for the presence of metastases. Manuals and Resources Note: All samples were collected into sterile tubes without anticoagulants. Imprecision was further examined to determine functional sensitivity Figure 1.
Moreover, the inability to detect malignant cells or other pathologic cell types eg, cell clusters and extracellular and intracellular bacteria strongly justifies the need for sustained visual morphologic examination of cytocentrifuged preparations. View large Download slide. Description Features Photos 1.
Inquire Here Reagent Type: Table 4 summarizes the clinical performance characteristics of the NC count for the different body fluids with their respective cutoff values for normality. Performance evaluation of the application of body fluids on the Sysmex XE series automated haematology analyzer.
Two Extraordinary Sellar Neuronal Tumors: On the other hand, the Cell-Dyn Sapphire failed to flag the remaining 8 samples with metastatic cells. Figure 1 shows the imprecision CV as a function of the cell count.
Please could you kindly explain in brief the differences between the two Machines In the present article, we describe its applicability for the distinct body fluids, and we highlight limitations and caveats. The analyzer features patented MAPSS technology plus 3 color fluorescence to provide our most comprehensive testing capability.
Cell Dyn Number of Assays: However, manual hemacytometer counts could even be more imprecise at low cell concentrations. BMononuclear cells with the cytocentrifuged preparation possibly including epithelial cells and macrophages.
[Evaluation of the Abbott Cell-Dyn Sapphire hematology analyzer].
Receive exclusive offers and updates from Oxford Academic. Receive your quote directly from the manufacturer. The poor reproducibility at low cell concentrations that we encountered is most likely to be held responsible for the absence of agreement with the manual microscopic count. Within-run imprecision of the microscopic NC count was also investigated at low cell counts.
Inquire Here Optical Source: Request Pricing Receive your quote directly from the manufacturer.